1. Remove the cap from the lancet device. 2. Insert a new lancet in the lancet holder. It is properly positioned when it snaps into place. To remove the protective cap, twist and pull it away from the lancet. 3. Replace the cap onto the lancet device. Twist the cap of the lancet device to set the lancet depth. Start at 2 or 3. For tougher sin, dial to a higher number. 4. Press the plunger on the end of the device all the way down until it clicks. A yellow dot will appear and remain in the center of the clear release button. 5. Wash your hands in warm, soapy water. Dry well. 6. Hold the lancet device against the side of your fingertip. 7. Press the release button. Gently squeeze the area until you get a small sample of blood. 8. Apply the drop of blood to the target area of your test strip. 9. Immediately remove used lancet. To avoid accidental injury, do not store lancet in device.
Do not use if protective guard has been previously removed from lancet. Do not reuse lancets or store lancets in device. This notice reminds the purchaser that a restricted patent license prohibits the use of Accu-Chek Softclix and Accu-Check Softclix Plus lancet devices with any lancet other than Accu-Check Softclix lancets.